Patients with severe Rheumatoid Arthritis may have significant ulnar drift of the digits which greatly interferes with their day-to-day activities. MCP arthroplasty is a procedure that may help realign the fingers to allow for improved function.
Indications for the surgery are volar subluxation and dislocation of the MCP joints with significant ulnar drift of the digits.
The goals of this surgery are to provide pain relief and to improve functional outcomes by realignment of the digits and correction of the ulnar drift and the extension deficits.
After removal of the metacarpal heads, the MCP joints are replaced by implants (usually made from silicone) with stems placed in the corresponding metacarpal and proximal phalanx. The implants help with correction of the deformity and allow for improved flexion and extension at the MCP joints.
Post-operatively the MCP joints are positioned in extension for about two weeks before any active therapy programme begins.
After suture removal, a thermoplastic orthosis is fabricated on the dorsal wrist and forearm and dynamic outriggers are created to support the MCP joints in extension while allowing for active MCP joint flexion. Care must be taken to create the proper angle of pull on the MCP joints. It may be necessary or recommended to add a radial based pull to counteract the deforming forces. The orthosis is to be worn for 4-6 weeks as the patient regains active motion in the fingers.
Summary of the objectives:
- Create support for the MCP joints in extension.
- Allow for active MCP joint flexion.
- Create a proper angle of pull on MCP joints.
- Ad radial based pull to counteract deforming forces (if needed).
- For 4 to 6 weeks.
I recently had the opportunity to assist with fabrication of this orthosis for a patient seen post MCP Arthroplasty at a clinic in New York City (U.S.).
Here are some key focus points:
The thermoplastic base orthosis can easily be fabricated from a variety of different materials. I prefer the properties of Orfilight, Orfilight Black NS and Orfilight Atomic Blue NS (3.2mm / 1/8”) as these materials are highly conforming, yet lightweight and comfortable.
The Orfitube adjusters are placed in alignment with the metacarpal bones on the dorsum of the hand.
The end caps are placed on the distal end of the Orfitube to prevent shearing forces on the elastic thread and they offer a finished and modern look.
The individual custom finger cuffs can be made from Orficast (3cm/1”). Activate a 2-3 inch (5 – 7,5 cm) piece of Orficast and lay the elastic thread near the bottom edge. Fold over the material so the thread is in the middle and mould around each finger’s proximal phalanx. Do not place the elastic thread into the fold, as this creates a lopsided support.
Follow the photos to create this fabrication:
Patient seen 10 days post-surgery. Notice severe ulnar drift on the right, with realigned fingers on the left.
Lining up all four adjusters with Orfitubes.
Checking placement and length of Orfitubes.
Check alignment from all angles.
The completed orthosis can be adjusted for increased radial pull as needed.
We added an Orficast buddy strap between the index and middle finger to help prevent rotation and overlapping of the fingers.
Written by Debby Schwartz, OTD, OTR/L, CHT
Physical Rehabilitation Product and Educational Specialist at Orfit Industries America.
Debby is a certified hand therapist with over 36 years of clinical experience. She completed her Doctorate of Occupational Therapy at Rocky Mountain University of Health Professions in 2010. She has worked at Orfit Industries America as Product and Educational Specialist since 2007.
Debby is also an adjunct professor at the Occupational Therapy Department of Touro University, School of Health Sciences, and at the Occupational Therapy Department at Yeshiva University, Katz School of Science and Health in NYC. She has written many book chapters in the field of hand therapy and multiple articles for hand therapy journals, including the ASHT Times and the Journal of Hand Therapy. She has published a new textbook on orthotic fabrication together with Dr. Katherine Schofield, entitled “Orthotic Design and Fabrication for the Upper Extremity: A Practical Guide”.
Contact us for more product information or find your local distributor here.
If you’d like to receive the latest product updates and interesting Orfit news, subscribe to our newsletter: